2-33. CONTROL BOX ASSEM BLY (CONT).
Do not bend or kink capillary tube. Any damage requires
replacement. Do not coil capillary tube in a coil less than six
inches in diameter.
Remove two nuts (20) from the rear of water temperature indicator ther-
mometer (21). Slide metal bracket (22) back onto capillary tube. Position
metal bracket (22) nearly parallel with the capillary tube. Carefully work
water temperature indicator thermometer (21) with metal bracket (22) out
through the mounting hole in the control box assembly (9).